Recall Posting Date
Surgical Duotome Sidelite Fibre for Holmiums
Recall Start Date
Model or Catalog #
Lot or Serial #
3912, 4011, 4116
Reason for Recall
1) Absence of black indicator markings on the metal tip and 2) fiber degradation (degradation of fiber, detachment of metal cap, and fiber lasing straight) during use for devices manufactured after Feb.1
"There's almost no risk involved," says Stephen Fanning, CEO of Solta Medical, which owns Fraxel.
The world is a dangerous place to live, not because of the people who do evil, but rather those who sit back & let it happen.